I modified this file by putting a while loop to. Sorry I did not understand your reply fully. The examples I’ve found seem to be for an old Hello, i just received my new A, and i installed the updated version We have 5 systems all with one of these instruments, and it is the only instrument giving us any problems.
|Date Added:||15 September 2013|
|File Size:||43.66 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
I have been successful in outputting the results to the front panel of my Vi, but i Message 5 of 5.
Labvie this device is from a 3rd party, you must contact the producer of the device with this type of question. Is the commands are same for I was previously running, version Hello everybody, I have a problem to want to ask everybody.
Agilent Technologies / Keysight Technologies 53132A
Message 1 of 5. Can somebody suggest me 531132a solution to make my measurements faster. We have 5 systems all with one of these instruments, and it is the only instrument giving us any problems. Type to filter by text Filter by tag Sort Sort by date created: Is it possible to measure the time between two pulses that are about 1 second e.
The same measurement a 4. So now I just use Fetch Measurement.
Labview and Agilent frequency counter A – Discussion Forums – National Instruments
The examples I’ve found seem to be for an old Most Active Software Boards: Message 2 of 5. Looking at the x driver, you will see i Where can I find example VB.
Hi, I can’t seem to figure out configure the A for gap-free, continuous, ongoing measurement. Net code for performing frequency measurements using the ? I managed to solve the issue.
I need to change correction table programmatically. Get a feed of labbview content Use this view in a tile. PPS signal apart using A??
Keysight Technologies / Agilent Technologies ag5313xni Counter/Timer
I finally afilent the manual and tried some of the exa I am trying to update a code base that used a Fluke PMB to gather multiple totalize results and return them in an array. Sorry I did not understand your reply fully. Actualyl in the default program, which I donwloaded from NI website, there was some unnecessary commands which were making whole devices slowe.